A whole family of solution to properly care and maintain your testers and meters.
These solutions are used to calibrate pH meters to assure they are accurate and working properly. You should check your pH meter frequently because the properties of the meters membrane is constantly changing from normal use and aging. We recommend a two point calibration between 4.0 and 7.0 to assure that your meter is accurate across the scale. This is why the Genesis solution is available in twin packs of 4.0 and 7.0. Ask your dealer for PHB-T for the twin pack available in 8oz or 16oz sizes.
PH Electrode Storage Solutions
Use to store your pH meter electrode between use. This solution is designed to keep your electrode sensing membrane and reference junction clean and unclogged and ready for use. Store you electrode directly in a small cup of solution or saturate a sponge in the cap of your meter for storage between use. This will prevent your permeable glass membrane from becoming dry and from mineral deposits clogging the surface. This solution is designed to equalize the osmotic pressure of the reference solution in your meter junction to keep the reference solution from seeping out and prematurely aging your meter. Should your electrode membrane dry out soak it in this solution for at least two hours before use. NEVER STORE AN ELECTRODE IN DISTILLED OR DEIONIZED WATER. Ask your dealer for Genesis ESS solution.
TDS and EC Standards
The Genesis Lab manufactures calibration standards for Total Dissolved Solids and Electrical Conductivity for the purpose of keeping your solution concentration meters reading properly. We recommend a two point calibration between 800 ppm tds (1200 uS) and 1300 ppm tds (2000 uS) to assure that your meter is accurate across the scale. This is why the Genesis solution is available in twin packs of 800 and 1300. Ask your dealer for PPM-T for the twin pack available in 8oz or 16oz sizes.
Also available in 1500 ppm tds (2.14 mS at 0.7 conv.), 1413 uS, 5000uS, and 12.88 mS. Other values available upon request.
Electrode and Probe Cleaning Solutions
Used to clean meter tip. Works for both pH and conductivity meters. Comes as a two part A and B cleaning solution to remove both organic and inorganic deposits from meter sensors. Use monthly or as needed to refresh your meter and keep it working optimum. Soak tip in part A for 5 min, then part B for 5 min and then back in part a for 5 more minutes and rinse with tap water.